Whys is it worth?:
Maintenance of laminar as devices ensuring the highest possible purity class A, the laminar-flow cabinets require cyclical specialist tests to prove their uninterrupted readiness to clean the air of impurities.Together with impurities, dusts, bacteria, viruses and fungi are removed. This has a direct impact on the safety of work with a given material, purity of the sample and the finished product.
Need for testing:
There are many factors that can lead to the so-called “leakage” of the filter, including the change of seal pressure, mechanical damage caused by the user or external factors.PN-EN IS 14644-2:2005 Part 2 specifies the maximum time between measurements as 24 months. This is also the period that we adopt for the newly-installed filters, further tests should be carried out more frequently – every 12 months.
It is necessary to show that the filter is still able to provide the desired purity class despite the continuing pollution.
The main parameters that are checked during measurements are:
We also test:
Additionally, we carry out of the relative humidity and temperature test.
HEPA filters – highly efficient air filters are manufactured and classified according to PN-EN 1822:2001. They are tested and classified on the basis of total effectiveness (99.995% for H14 filter) and local effectiveness (99.75% for H14 filter) for the most penetrating particles (MPPS).
HEPA filter life is estimated at 3,000 hours or 5 years (whichever comes sooner), but due to different working conditions, the hourly limit of the filter cannot be accurately determined.Our filters come from reliable manufacturers with long-standing experience.
Our Company has over 20 years of experience in the production of laminar-flow cabinets used not only in pharmacies and laboratories (Safety Class I), but also in oncology and bacteriology (Safety Class II). We have the knowledge and experience in the field of testing, filter replacement and service of laminar-flow cabinets, also cabinets produced by other manufacturers.You can use our maintenance services throughout the country.
Our control devices comply with all standards and have a valid calibration certificate.
Each maintenance service is completed with the issuance of a relevant report.
Maintenance of laminar services:
The original certificate is valid for 2 years of operation or a new one when the filter was replaced.In the 2nd, 3rd and 4th year of operation, it is required to make a control check.
In the 5th year of operation, HEPA filter should be replaced and control check completed with a 2-year certificated is required.
Regardless of the passage of time, it should be replaced after 3,000 hours of operation.